Measurement resolution of 0.02°C
Temp Range: -20 to 80°C
Sensing Type: Infrared
Audio Output: 3 Watts
DC Input Voltage: 9V DC, 2 Amps
AC Input Voltage: 220V ~ 0.3Amp
To ensure the accuracy of measurement and safety of use, please read the manual carefully before use.
1 Product Introduction
The infrared thermometer is a high-quality product, it adopts infrared technology, each start-up will be self-testing to ensure the accuracy of measurement, this infrared thermometer is mainly used to measure the temperature of the forehead of the human body. This product can perform accurate and stable temperature measurement. Users only need to point the probe head on the forehead and press the measurement button, and the body temperature can be measured quickly and accurately in one second. To ensure the accuracy of measurement and safety of use, please read the manual carefully before use.
This product is widely used in schools, customs, hospitals, stadium, Shopping Mall, Bus Counter, homes and other places.
The body temperature is measured by measuring the human body's thermal radiation.
3 Structure and Composition
It consists of a housing, an infrared temperature sensor, a signal receiving processor, buttons, a COB board, a buzzer, and an LCD display. 4 Preventive Examination
a. Before measurement, the infrared thermometer shall be proofread to check the function of each component and keep the good performance of the infrared thermometer.
b. Maintain sufficient power supply to determine whether the external environment is in accordance with the product operating environment.
c. After the product is packed, it is allowed to be transported by common means of transportation, but it should avoid the mechanical collision of rain, damp and extrusion box.
This product is intended for use only as described in the instructions. The manufacturer shall not be liable for any damage caused by improper use. I. Do not immerse this product in water or other liquids. For cleaning, wipe with a damp soft cloth.
II. If you think this product is damaged or abnormal, please stop using it.
III. Do not open this product without permission.
IV. The blood vessels constrict and the skin surface temperature drops during the early stages of fever, when the measured temperature is unusually low.
V. If the measurements do not match the patient's diagnosis or the temperature is abnormally low, the measurements are repeated every 15 minutes or another core temperature area is measure to verify the previous measurements.
VI. This equipment includes sensitive elements and shall be treated with caution. The storage and operating conditions are described in the Product Specification section.
VII. This product contains small parts, in order to avoid swallowing, children need to use under adult supervision.
** Please stay away from the following situations:
a) Extreme temperature
b) Shock and drop
c) Pollution and Dust
d) Direct sunlight
e) Hot and cold environment
Warning: This device is not a complete substitute for medical treatment. It is not waterproof and shall not be used in liquids.
Warranty and After Sales Support
1. This product is used for five years from the date of purchase; you can enjoy one-year free warranty.
2. The packaging is not covered by the warranty.
3. For the following damage caused by the user, please forgive us for not providing free warranty service;
a) Failure caused by unauthorized disassembly and modification.
b) Failure caused by accidental drop during use or handling;
c) Failure caused by not following the correct instructions in the manual;
d) Failure caused by lack of reasonable maintenance;
e) When requesting to provide free warranty service, you must hold the warranty card filled with the date of purchase and the seal of the purchase dealer (including the name and address of the dealer).
f) Repair services outside the warranty scope will be charged according to the corresponding regulations;
g) When requesting free warranty service, please take this product to the dealers of our company for repair.
• The protective glass covering the lens is the most fragile part of the thermometer.
Please take great care to prevent damage.
• Do not use rechargeable batteries.
• Do not place device in fire.
• Do not expose the device to sunlight or water.
• Maintain measuring distance of 2cm
• Prevent damage to the infrared sensor.
• Clean the device surface with a cotton pad lightly moistened with 70% Isopropyl alcohol.
• Do not clean with corrosive detergent.
• Keep the device away from water or other liquid.
• Store in a dry environment.
• Keep away from dust and direct sunlight.
KEEP OUT OF REACH OF ALL CHILDREN
Product specifications subject to change without notice
For the latest version of this User’s Guide, Software updates, and other up-to-the-minute product information, visit our website: www.hawkish.co.uk
Manufactured by: Frank Volacrest
Made in Bangladesh
All rights reserved including the right of reproduction in whole or in part in any form.